allows you to quickly verify the health of the quartz resonators. Diagram of the device consists of a generator (T1), detector (D1, D2) and DC amplifier (T2). By connecting the quartz to the two terminals of the generator, the power on. If the resonator is OK, resistor R2 appears to the high-frequency voltage which is then fed to diodes D1, D2 for detection. Highlighted in this case a constant component opens the transistor amplifier stage. Load UPT serve milliammeter ИП1 and lamp L1. The glow lamp indicates the functionality of the quartz. But his activity is judged by the readings of the milliammeter. For the active resonator current is 70-90 mA, and for inactive — 30-40 mA.
Test quartzes can be powered from the battery voltage 9-12 V, which is connected instead of the rectifier.
Schematic diagram of the test quartzes.
Data details are given of the PA scheme. Capacitors C1, C3, C4 — CBC, C2 — CTC, C5 — C50-6. Milliammeter type M5-2 with the current full deflection of 100 mA. As Tr 1 used television unified the transformer TCE-70. L1 — switch lamp 12 ВХ60 mA.
G. VEREIN, Ordzhonikidze
Noticed mistape? Highlight it and press Ctrl+Enter, to inform us.
Recommend to read
- WITHOUT ALTERATION OF THE POWER SUPPLY
In Amateur radio practice, there are situations when 9-volt DC source you need to obtain 12, and then a higher output voltage. Bail in such cases can I designed a compact transformerless...
- BATTERY UNDER CONTROL
Any motorist knows that the state of charge of the battery must constantly monitor is required to control the pressure on it. This can be done by using electronic circuit which gives an...